CMS5000 Patient Monitor adopts Oscillometry technology, also called Oscillography technology for NIBP measuring, and adopts Photoelectric Oxyhemoglobin Inspection Technology combining Capacity Pulse Scanning & Recording Technology for SpO2 measuring. Systolic Pressure (SYS), Diastolic Pressure (DIA), Mean Pressure (MAP), SpO2 and Pulse rate (PR) can be accurately measured.
Compact appearance, comprehensive functions, simple and convenient operation, which is applicable for hospitals, community medical treatment and family.
Be applicable for medicine, surgery, operating room, ICU/CCU, emergency room, obstetrics and gynecology, pediatrics.
Built-in rechargeable lithium polymer battery for uninterrupted monitoring.
Compact and flexible appearance, easy for carrying and be suitable for indoor and outdoor(in ambulance) monitoring.
With user-friendly interface.
Display with 2.4'' (320×240) color TFT LCD and blue highlight LED.
Visual and audible alarm for Systolic Pressure(SYS), Diastolic Pressure(DIA), Mean Pressure(MAP), SpO2 and Pulse rate(PR), and upper and lower limit of alarm can be set.
Unique Flash memory, storage for up to 2000 NIBP data or SpO2 data within 48-hour.
Convenient and quick in inquiring measurement data.
Optional parameter: temperature.
Measurement method: Oscillometry
Measurement mode: manual/auto.
Auto measurement interval: adjustable from 1 to 255 minutes.
Accuracy: Maximum Mean deviation±5mmHg, Maximum Standard deviation≤8mmHg.
Overpressure protection: double protection for software and hardware.
2.Upper and lower limits of alarming
Continuous real-time measuring.
Adopt digital SpO2 technology, which has strong anti-interference capability.
Real-time display SpO2, pulse rate, SpO2 Plethysmogram, bar graph, SpO2 and Pulse rate (PR) trend graph.
Trend graph with the function of adaptive scaling
Standard adult finger-clip probe, binding probe for neonate is optional.
Range: 0%~100% (Resolution: 1%)
Accuracy: ±2% in stage of 70%-100%, and unspecified in stage of less than 70%.
2.Pulse Rate Measuring
Range: 30bpm~250bpm(Resolution: 1bpm)
Accuracy: ±2bpm or ±2%, whichever is greater.
3.Measurement Performance in Weak Filling Condition: SpO2 and pulse rate can be shown correctly when pulse-filling ratio is 0.4%, SpO2 error is ±4%; the Pulse rate error is ±2bpm or ±2%, whichever is greater.
4.Resistance to surrounding light: The deviation between the value measured in the condition of man-made light or indoor natural light and that of darkroom is less than ±1%.
Adult BP cuff
Extending tube for blood pressure
User Manual （GB only）on request for other Language
Dimension: 232 mm(L)×198 mm(W)×75 mm(H)